### update

parent b1d0cccd
 ... ... @@ -326,10 +326,6 @@ two ADS1115 were used,one for the voltage measurement one for the current measur is a 16-bit ADC (Analog-to-Digital Converter), with an adaptable gain. The advantage of ADS1115 is that the input signal value could lie between - to + 6.114 V. For the current measurement we have directly integrated the INA282 component, which allows to realize precise current measurement around a shunt resistor. The assembly are described in the following steps:

Note

In this version, We used a shunt resistor of 2 ohms, which limits the current measurement to 48 mA. If the current is higher than this value, you just have to decrease the value of the shunt resistor.

... ... @@ -487,6 +483,10 @@ electrodes A and B

Warning

In this version, We used a shunt resistor of 2 ohms, which limits the current measurement to 48 mA. If the current is higher than this value, you just have to decrease the value of the shunt resistor.Change the shunt value in the code.

... ...
 ... ... @@ -43,9 +43,6 @@ input signal value could lie between - to + 6.114 V. For the current measurement which allows to realize precise current measurement around a shunt resistor. The assembly are described in the following steps: .. note:: In this version, We used a shunt resistor of 2 ohms, which limits the current measurement to 48 mA. If the current is higher than this value, you just have to decrease the value of the shunt resistor. .. table:: :align: center ... ... @@ -155,6 +152,12 @@ which allows to realize precise current measurement around a shunt resistor. The | | Soldering three capacitors (100nF) | | | | +---------+------------------------------------------------------------+ .. warning:: In this version, We used a shunt resistor of 2 ohms, which limits the current measurement to 48 mA. If the current is higher than this value, you just have to decrease the value of the shunt resistor.Change the shunt value in the code. .. table:: :align: center ... ...
This diff is collapsed.
No preview for this file type
No preview for this file type
 ... ... @@ -326,10 +326,6 @@ two ADS1115 were used,one for the voltage measurement one for the current measur is a 16-bit ADC (Analog-to-Digital Converter), with an adaptable gain. The advantage of ADS1115 is that the input signal value could lie between - to + 6.114 V. For the current measurement we have directly integrated the INA282 component, which allows to realize precise current measurement around a shunt resistor. The assembly are described in the following steps:

Note

In this version, We used a shunt resistor of 2 ohms, which limits the current measurement to 48 mA. If the current is higher than this value, you just have to decrease the value of the shunt resistor.

... ... @@ -487,6 +483,10 @@ electrodes A and B

Warning

In this version, We used a shunt resistor of 2 ohms, which limits the current measurement to 48 mA. If the current is higher than this value, you just have to decrease the value of the shunt resistor.Change the shunt value in the code.

... ...
 ... ... @@ -43,9 +43,6 @@ input signal value could lie between - to + 6.114 V. For the current measurement which allows to realize precise current measurement around a shunt resistor. The assembly are described in the following steps: .. note:: In this version, We used a shunt resistor of 2 ohms, which limits the current measurement to 48 mA. If the current is higher than this value, you just have to decrease the value of the shunt resistor. .. table:: :align: center ... ... @@ -155,6 +152,12 @@ which allows to realize precise current measurement around a shunt resistor. The | | Soldering three capacitors (100nF) | | | | +---------+------------------------------------------------------------+ .. warning:: In this version, We used a shunt resistor of 2 ohms, which limits the current measurement to 48 mA. If the current is higher than this value, you just have to decrease the value of the shunt resistor.Change the shunt value in the code. .. table:: :align: center ... ...
This diff is collapsed.
 ... ... @@ -43,9 +43,6 @@ input signal value could lie between - to + 6.114 V. For the current measurement which allows to realize precise current measurement around a shunt resistor. The assembly are described in the following steps: .. note:: In this version, We used a shunt resistor of 2 ohms, which limits the current measurement to 48 mA. If the current is higher than this value, you just have to decrease the value of the shunt resistor. .. table:: :align: center ... ... @@ -155,6 +152,12 @@ which allows to realize precise current measurement around a shunt resistor. The | | Soldering three capacitors (100nF) | | | | +---------+------------------------------------------------------------+ .. warning:: In this version, We used a shunt resistor of 2 ohms, which limits the current measurement to 48 mA. If the current is higher than this value, you just have to decrease the value of the shunt resistor.Change the shunt value in the code. .. table:: :align: center ... ...
Markdown is supported
0% or .
You are about to add 0 people to the discussion. Proceed with caution.
Finish editing this message first!